# Changes between Version 6 and Version 7 of WikiStart

Ignore:
Timestamp:
Sep 8, 2017, 5:46:59 PM (3 years ago)
Comment:

--

### Legend:

Unmodified
 v6 = 4 - Methods of calibration and data Processing (NADINE) = ''The MSCTI conductivity probe is calibrated after each set of experiments when the tank is drained (see Section 3.1 for full details). The ADVs have 4 heads and as such this enables some internal verification of the instrument. The ADV and UVP datasets are processed using a series of bespoke Matlab scripts. The PIV data will be processed using a bespoke script. Access to commercial PIV processing packages is also available.'' ''The images for PIV are calibrated from images of grid put in 0_REF_FILES/Calib_absolu. The 3D calibration involves 'intrinsic parameters' of the optical system obtained from images of the same grid seen with tilt angle (put in /Calib-14-09-3D). Then rotations and translations of the calibration points are introduced to adjust the relationship between image coordinates and physical coordinates defined in section 2.2. See http://servforge.legi.grenoble-inp.fr/projects/soft-uvmat/wiki/UvmatHelp#GeometryCalib for details of the method. The calibration parameters are copied in a xml file beside each image folder with the same name (for instance PCO2.xml for PCO2/). The xml files also containing all the timing information.'' ''All the images and processing results from the images are in the folder 0_PIV under the folder with the name of the experiment.'' ''The images are first extracted from their initial format and written as .png images (compression with no loss of data) labelled by two indices i and j=1 to 20. The index i generally runs from 1 to 150 scanning 15 levels (then coming back to each level 1O times).'' ''A first step in image processing after extraction is to subtract the fixed background and rescale the image intensity leading to a image folder with extension .sback. PIV results are stored as netcdf files (extension .nc) in a folder .sback.civ. These data are still in pixel displacement.'' ''Final velocity data in phys coordinates are stored as 2D matrices under the netcdf format in folders with extension .sback.civ.mproj. They are defined on a physical grid with 1 cm mesh.'' The PIV data will be processed using the MATLAB tool UVMAT available  on !http://servforge.legi.grenoble-inp.fr/projects/soft-uvmat. All  images from the experiments will be stored in the directory  /fsnet/projects/coriolis/2017/17ICESHELF/DATA under the name for the  experiment. == '''4.1 Calibration for shelf break experiments''' == The images for PIV are calibrated from images of a horizontal grid and in 3D with an inclined grid (tilted angles). [[BR]]The calibrated images with the grids are stored in 0_REF_FILES with the [[BR]]- 2D calibration in the directory CALIB_07_09[[BR]]- 3D calibration in the directory CALIB_07_09_3D[[BR]]The images are divided into the two different cameras as PCO1/ and PCO2/. The  3D calibration is done with the geometric calibration function on UVMAT  to create 'intrinsic parameters' for each camera that will be used for  all images. Calibration points along the inclined grid are marked and  translated from image coordinates to physical coordinates defined in  section 2.3.1 to later take into account the different height of slices. See http://servforge.legi.grenoble-inp.fr/projects/soft-uvmat/wiki/UvmatHelp#GeometryCalib  for details of the method. The calibration parameters are copied in a  xml file beside the folders of each camera (for instance PCO2.xml for  PCO2/). The xml files also containing all the timing information and  have to be copied next to the folders of all images, to use the same  intrinsic calibration data for each image taken with the corresponding  camera. The 2D calibration is used as a reference horizontal grid with a grid height of[[BR]]- z = 80.8 cm for PCO1[[BR]]- z = 58.2 cm for PCO2 The  calibration is checked with a calibration ruler (images saved in  CALIB_RULER) to see that the transition between the two cameras is  smooth. == 4.2 Calibration for ice shelf experiments == = 5 - Organization of data files = ''All data related to the project are in: \\SERVAUTH4\share\project\coriolis\2016 or \\servauth4.legi.grenoble-inp.fr\share\project\coriolis\2016'' = ''6 - Table of Experiments:'' = ||'''''Exp No.'''''||'''''Name'''''||'''''Date_begin'''''||'''''$H_{total}'''''||'''''$Q$'''''||'''''f'''''||'''''R_{curvatture}'''''||'''''\Delta \rho'''''||'''''A_flap'''''||'''''Vmax_flap'''''||'''''Q_jet'''''||'''''d_jet'''''||'''''dt_forcing'''''||'''''delay'''''||'''''dt_record'''''||'''''Cameras'''''|| ||''''||''''||''''||''$(cm)$''||''$(m)$''||''$s{-1}$''||''$(°)$''||''$(s)$''||''$(cm)$''||''$(cm/s)$''||''$(l/min)$''||''$(cm)$''||''$(s/min)$''||''$(s)$''||''$(s)$''||''$names$''|| ||'''''Test Exp'''''||''''||''''||''''||''''|| ||'''''''||'||'''''''||''$(cm)$''||''$(m)$''||''$s{-1}$''||''$(°)$''||''$(s)$''||''$(cm)$''||''$(cm/s)$''||''$(l/min)$''||''$(cm)$''||''$(s/min)$''||''$(s)$''||''$(s)$''||''$names$''|| ||'''''Test Exp'''''||'''''''||'||'''''''||'|| ||''1''||''EXP01''||''??????''||''17.7''||''1''||''60''||''90''||''200''||''60''||''0.9''||''-''||''-''||''2000 / 34''||''0''||''9856.5''||''PCO''|| ||''2''||''EXP02''||''??????''||''17.7''||''1''||''60''||''90''||''100''||''60''||''1.9''||''-''||''-''||''3000 / 50''||''100''||''3250''||''PCO''|| ||''9''||''EXP09''||''??????''||''17.6''||''1''||''-''||''-''||''-''||''-''||''-''||''7.4''||''60''||''1336.5/23''||''0''||''1336.5''||''PCO''|| ||''10''||''EXP10''||''??????''||''17.6''||''1''||''-''||''-''||''-''||''-''||''-''||''35.3 to 60''||''60''||''2965.5/49''||''0''||''2965.5''||''PCO''|| ||''''||''''||''''||''''||''''||''''||''''||''''||''''||''''||''''||''''||''''||''''||''''|| ||''''||''''||''''||''''||''''||''''||''''||''''||''''||''''||''''||''''||''''||''''||''''|| ||'''''Exp Canyon'''''||''''||''''||''''||''''|| ||'''''''||'||'''''''||'||'''''''||'||'''''''||'||'''''''||'||'''''''||'||'''''''||'||'''''''|| ||'''''''||'||'''''''||'||'''''''||'||'''''''||'||'''''''||'||'''''''||'||'''''''||'||'''''''|| ||'''''Exp Canyon'''''||'''''''||'||'''''''||'|| ||''11''||''EXP11''||''22/02 11:41''||''18''||''0''||''0''||''90''||''60''||''16''||''0.8''||''-''||''-''||''3000/50''||''50''||''3300''||''PCO/FALCON/Lumenera''|| ||''12''||''EXP12''||''22/02 15:33''||''18''||''0''||''0''||''90''||''60''||''22''||''1.2''||''-''||''-''||''3000/50''||''300''||''3300''||''PCO/FALCON/Lumenera''|| ||''31''||''EXP31''||''03/03 14:42''||''18''||''0''||''0''||''-''||''-''||''-''||''-''||''120''||''60''||''???''||''0''||''???''||''PCO/FALCON/Lumenera''|| ||''32''||''EXP32''||''03/03 16:08''||''18''||''0''||''0''||''-''||''-''||''-''||''-''||''75''||''60''||''???''||''0''||''???''||''PCO/FALCON/Lumenera''|| ||''''||''''||''''||''''||''''||''''||''''||''''||''''||''''||''''||''''||''''||''''||''''|| ||'''''Exp Smooth Shelf'''''||''''||''''||''''||''''|| ||'''''''||'||'''''''||'||'''''''||'||'''''''||'||'''''''||'||'''''''||'||'''''''||'||'''''''|| ||'''''Exp Smooth Shelf'''''||'''''''||'||'''''''||'|| ||''33''||''EXP33''||''08/03 14:55''||''19''||''0''||''0''||''-''||''-''||''-''||''-''||''37.5''||''60''||''???''||''0''||''???''||''PCO/FALCON/Lumenera''|| ||''34''||''EXP34''||''08/03 16:26''||''19''||''0''||''0''||''-''||''-''||''-''||''-''||''69''||''60''||''???''||''0''||''???''||''PCO/FALCON/Lumenera''|| '''''EXP01''': '''Oscillation''' period 200s, very poor repartition of particles and hard to distinguish the global flow structure.'' ||'''''Cameras'''''||'''''location'''''||'''''exposure'''''||'''''dt'''''||'''''size'''''||'''''avg scale'''''||''''|| ||''''||''''||''$(ms)$''||''$(s)$''||''$pix.pix$''||''$pix/cm$''||''''|| ||''PCO''||''after carriage''||''100ms''||''0.1''||''?''||''?''||''''|| ||'''''Cameras'''''||'''''location'''''||'''''exposure'''''||'''''dt'''''||'''''size'''''||'''''avg scale'''''||'''''''|| ||'''''''||'||''$(ms)$''||''$(s)$''||''$pix.pix$''||''$pix/cm$''||'''''''|| ||''PCO''||''after carriage''||''100ms''||''0.1''||''?''||''?''||'''''''|| ''Laser parameters'' ||'''''Laser'''''||'''''Intensity'''''||'''''angle'''''||'''''d-3'''''||''''||'''''d-2'''''||'''''d-1'''''||'''''d0'''''||'''''d1'''''||'''''d2'''''||'''''d3'''''|| ||''$(nm)$''||''$(Watt)$''||''$(°)$''||''$(cm)$''||''''||''$(cm)$''||''$(cm)$''||''$(cm)$''||''$(cm)$''||''$(cm)$''||''$(cm)$''|| ||''532''||''25''||''60''||''''||''''||''''||''''||''''||''''||''''||''''|| ||'''''Laser'''''||'''''Intensity'''''||'''''angle'''''||'''''d-3'''''||'''''''||''d-2''''''''||''d-1''''''''||''d0''''''''||''d1''''''''||''d2''''''''||''d3''''''''|| ||''$(nm)$''||''$(Watt)$''||''$(°)$''||''$(cm)$''||'''''''||''$(cm)$''||''$(cm)$''||''$(cm)$''||''$(cm)$''||''$(cm)$''||''$(cm)$''|| ||''532''||''25''||''60''||'''''''||'||'''''''||'||'''''''||'||'''''''||'|| '''''EXP02''': '''Oscillation''' period 100s, we starts to see some oscillation in the flow that propagates along the wall. A large-scale shear flow seems to establish along the shelf.'' ||'''''Cameras'''''||'''''location'''''||'''''exposure'''''||'''''dt'''''||'''''size'''''||'''''avg scale'''''||''''|| ||''''||''''||''$(ms)$''||''$(s)$''||''$pix.pix$''||''$pix/cm$''||''''|| ||''PCO''||''after carriage''||''50ms''||''0.5''||''?''||''?''||''''|| ||'''''Cameras'''''||'''''location'''''||'''''exposure'''''||'''''dt'''''||'''''size'''''||'''''avg scale'''''||'''''''|| ||'''''''||'||''$(ms)$''||''$(s)$''||''$pix.pix$''||''$pix/cm$''||'''''''|| ||''PCO''||''after carriage''||''50ms''||''0.5''||''?''||''?''||'''''''|| '''''EXP03''': '''Oscillation''' period 100s, reduced oscillation amplitude'' ||'''''Cameras'''''||'''''location'''''||'''''exposure'''''||'''''dt'''''||'''''size'''''||'''''avg scale'''''||''''|| ||''''||''''||''$(ms)$''||''$(s)$''||''$pix.pix$''||''$pix/cm$''||''''|| ||''PCO''||''after carriage''||''50ms''||''0.5''||''?''||''?''||''''|| ||'''''Cameras'''''||'''''location'''''||'''''exposure'''''||'''''dt'''''||'''''size'''''||'''''avg scale'''''||'''''''|| ||'''''''||'||''$(ms)$''||''$(s)$''||''$pix.pix$''||''$pix/cm$''||'''''''|| ||''PCO''||''after carriage''||''50ms''||''0.5''||''?''||''?''||'''''''|| '''''Thursday, February 16th 2017''''' '''''EXP04''': '''Oscillation''' period 60s'' ||'''''Cameras'''''||'''''location'''''||'''''exposure'''''||'''''dt'''''||'''''size'''''||'''''avg scale'''''||''''|| ||''''||''''||''$(ms)$''||''$(s)$''||''$pix.pix$''||''$pix/cm$''||''''|| ||''PCO''||''after carriage''||''50ms''||''0.1''||''?''||''?''||''''|| ||'''''Cameras'''''||'''''location'''''||'''''exposure'''''||'''''dt'''''||'''''size'''''||'''''avg scale'''''||'''''''|| ||'''''''||'||''$(ms)$''||''$(s)$''||''$pix.pix$''||''$pix/cm$''||'''''''|| ||''PCO''||''after carriage''||''50ms''||''0.1''||''?''||''?''||'''''''|| ''Laser parameters'' ||'''''Laser'''''||'''''Intensity'''''||'''''angle'''''||'''''d-3'''''||''''||'''''d-2'''''||'''''d-1'''''||'''''d0'''''||'''''d1'''''||'''''d2'''''||'''''d3'''''|| ||''$(nm)$''||''$(Watt)$''||''$(°)$''||''$(cm)$''||''''||''$(cm)$''||''$(cm)$''||''$(cm)$''||''$(cm)$''||''$(cm)$''||''$(cm)$''|| ||''532''||''25''||''75''||''''||''''||''''||''''||''''||''''||''''||''''|| ||'''''Laser'''''||'''''Intensity'''''||'''''angle'''''||'''''d-3'''''||'''''''||''d-2''''''''||''d-1''''''''||''d0''''''''||''d1''''''''||''d2''''''''||''d3''''''''|| ||''$(nm)$''||''$(Watt)$''||''$(°)$''||''$(cm)$''||'''''''||''$(cm)$''||''$(cm)$''||''$(cm)$''||''$(cm)$''||''$(cm)$''||''$(cm)$''|| ||''532''||''25''||''75''||'''''''||'||'''''''||'||'''''''||'||'''''''||'|| '''''EXP05''': '''Oscillation''' period 40s'' ||'''''Cameras'''''||'''''location'''''||'''''exposure'''''||'''''dt'''''||'''''size'''''||'''''avg scale'''''||''''|| ||''''||''''||''$(ms)$''||''$(s)$''||''$pix.pix$''||''$pix/cm$''||''''|| ||''PCO''||''after carriage''||''50ms''||''0.5''||''?''||''?''||''''|| ||'''''Cameras'''''||'''''location'''''||'''''exposure'''''||'''''dt'''''||'''''size'''''||'''''avg scale'''''||'''''''|| ||'''''''||'||''$(ms)$''||''$(s)$''||''$pix.pix$''||''$pix/cm$''||'''''''|| ||''PCO''||''after carriage''||''50ms''||''0.5''||''?''||''?''||'''''''|| '''''EXP06''': '''Oscillation''' period 80s'' ||'''''Cameras'''''||'''''location'''''||'''''exposure'''''||'''''dt'''''||'''''size'''''||'''''avg scale'''''||''''|| ||''''||''''||''$(ms)$''||''$(s)$''||''$pix.pix$''||''$pix/cm$''||''''|| ||''PCO''||''after carriage''||''50ms''||''0.5''||''?''||''?''||''''|| ||'''''Cameras'''''||'''''location'''''||'''''exposure'''''||'''''dt'''''||'''''size'''''||'''''avg scale'''''||'''''''|| ||'''''''||'||''$(ms)$''||''$(s)$''||''$pix.pix$''||''$pix/cm$''||'''''''|| ||''PCO''||''after carriage''||''50ms''||''0.5''||''?''||''?''||'''''''|| '''''EXP07''': '''Oscillation''' period 20s'' ||'''''Cameras'''''||'''''location'''''||'''''exposure'''''||'''''dt'''''||'''''size'''''||'''''avg scale'''''||''''|| ||''''||''''||''$(ms)$''||''$(s)$''||''$pix.pix$''||''$pix/cm$''||''''|| ||''PCO''||''after carriage''||''50ms''||''0.5''||''?''||''?''||''''|| ||'''''Cameras'''''||'''''location'''''||'''''exposure'''''||'''''dt'''''||'''''size'''''||'''''avg scale'''''||'''''''|| ||'''''''||'||''$(ms)$''||''$(s)$''||''$pix.pix$''||''$pix/cm$''||'''''''|| ||''PCO''||''after carriage''||''50ms''||''0.5''||''?''||''?''||'''''''|| ''At the end of the day, we want to perform an experiment with the laser in the upper layer but the upper limit of the laser is inside the layer of transition between the upper and lower layer. Consequently, it changes the angle of the laser due to the refraction of the beam into the inhomogeneous layer. This upper limit is fixed by the position of the mirror on which the laser beam is reflected that need to be moved when the tank is empty.'' '''''EXP08''': '''Injection''' flow rate 7.2 l/min'' ||'''''Cameras'''''||'''''location'''''||'''''exposure'''''||'''''dt'''''||'''''size'''''||'''''avg scale'''''||''''|| ||''''||''''||''$(ms)$''||''$(s)$''||''$pix.pix$''||''$pix/cm$''||''''|| ||''PCO''||''after carriage''||''50ms''||''0.5''||''?''||''?''||''''|| ||'''''Cameras'''''||'''''location'''''||'''''exposure'''''||'''''dt'''''||'''''size'''''||'''''avg scale'''''||'''''''|| ||'''''''||'||''$(ms)$''||''$(s)$''||''$pix.pix$''||''$pix/cm$''||'''''''|| ||''PCO''||''after carriage''||''50ms''||''0.5''||''?''||''?''||'''''''|| '''''EXP09''': '''Injection''' flow rate 7.4 l/min'' ||'''''Cameras'''''||'''''location'''''||'''''exposure'''''||'''''dt'''''||'''''size'''''||'''''avg scale'''''||''''|| ||''''||''''||''$(ms)$''||''$(s)$''||''$pix.pix$''||''$pix/cm$''||''''|| ||''PCO''||''after carriage''||''50ms''||''0.5''||''?''||''?''||''''|| ||'''''Cameras'''''||'''''location'''''||'''''exposure'''''||'''''dt'''''||'''''size'''''||'''''avg scale'''''||'''''''|| ||'''''''||'||''$(ms)$''||''$(s)$''||''$pix.pix$''||''$pix/cm$''||'''''''|| ||''PCO''||''after carriage''||''50ms''||''0.5''||''?''||''?''||'''''''|| '''''EXP10''': '''Injection''' flow rate 35.3 l/min to 60 l/min. We increased the flow rate from 35.3 l/min to 60 l/min after XXXX minutes of recording the see its impact on the formation of the coastal current.'' ||'''''Cameras'''''||'''''location'''''||'''''exposure'''''||'''''dt'''''||'''''size'''''||'''''avg scale'''''||''''|| ||''''||''''||''$(ms)$''||''$(s)$''||''$pix.pix$''||''$pix/cm$''||''''|| ||''PCO''||''after carriage''||''50ms''||''0.5''||''?''||''?''||''''|| ||'''''Cameras'''''||'''''location'''''||'''''exposure'''''||'''''dt'''''||'''''size'''''||'''''avg scale'''''||'''''''|| ||'''''''||'||''$(ms)$''||''$(s)$''||''$pix.pix$''||''$pix/cm$''||'''''''|| ||''PCO''||''after carriage''||''50ms''||''0.5''||''?''||''?''||'''''''|| '''''Monday, February 20th 2017''''' ||''Position''||''M2''||''M1''||''M0''||''M-1''||''M-2''|| ||''H_coast (cm)''||''''||''''||''18''||''''||''''|| ||''H_laser_up (cm)''||''''||''''||''''||''''||''''|| ||''H_laser_down (cm)''||''''||''''||''''||''''||''''|| ||''H_coast (cm)''||'''''''||'||''18''||'''''''||'|| ||''H_laser_up (cm)''||'''''''||'||'''''''||'||'''''''|| ||''H_laser_down (cm)''||'''''''||'||'''''''||'||'''''''|| '''''Wednesday, February 22th 2017''''' '''''EXP21''': '''Oscillation''' period 80s'' ||'''''Cameras'''''||'''''location'''''||'''''exposure'''''||'''''dt'''''||'''''size'''''||'''''avg scale'''''||''''|| ||''''||''''||''$(ms)$''||''$(s)$''||''$pix.pix$''||''$pix/cm$''||''''|| ||''PCO/FALCON/Lumenera''||''before carriage''||''50ms''||''0.3''||''?''||''?''||''''|| ||'''''Cameras'''''||'''''location'''''||'''''exposure'''''||'''''dt'''''||'''''size'''''||'''''avg scale'''''||'''''''|| ||'''''''||'||''$(ms)$''||''$(s)$''||''$pix.pix$''||''$pix/cm$''||'''''''|| ||''PCO/FALCON/Lumenera''||''before carriage''||''50ms''||''0.3''||''?''||''?''||'''''''|| [[BR]]'''''Monday, February 27th 2017''''' '''''EXP22''': '''Oscillation''' '''__Barotropic__''' period 60s amplitude 16cm'' ||'''''Cameras'''''||'''''location'''''||'''''exposure'''''||'''''dt'''''||'''''size'''''||'''''avg scale'''''||''''|| ||''''||''''||''$(ms)$''||''$(s)$''||''$pix.pix$''||''$pix/cm$''||''''|| ||''PCO/FALCON/Lumenera''||''before carriage''||''50ms''||''0.3''||''?''||''?''||''''|| ||'''''Cameras'''''||'''''location'''''||'''''exposure'''''||'''''dt'''''||'''''size'''''||'''''avg scale'''''||'''''''|| ||'''''''||'||''$(ms)$''||''$(s)$''||''$pix.pix$''||''$pix/cm$''||'''''''|| ||''PCO/FALCON/Lumenera''||''before carriage''||''50ms''||''0.3''||''?''||''?''||'''''''|| '''''EXP23''': '''Oscillation''' '''__Barotropic__''' period 60s amplitude 46cm'' ||'''''Cameras'''''||'''''location'''''||'''''exposure'''''||'''''dt'''''||'''''size'''''||'''''avg scale'''''||''''|| ||''''||''''||''$(ms)$''||''$(s)$''||''$pix.pix$''||''$pix/cm$''||''''|| ||''PCO/FALCON/Lumenera''||''before carriage''||''50ms''||''0.3''||''?''||''?''||''''|| ||'''''Cameras'''''||'''''location'''''||'''''exposure'''''||'''''dt'''''||'''''size'''''||'''''avg scale'''''||'''''''|| ||'''''''||'||''$(ms)$''||''$(s)$''||''$pix.pix$''||''$pix/cm\$''||'''''''|| ||''PCO/FALCON/Lumenera''||''before carriage''||''50ms''||''0.3''||''?''||''?''||'''''''|| ''The recycling of the water is started at the end of the day.'' ||''Position''||''M2''||''M1''||''M0''||''M-1''||''M-2''|| ||''H_coast (cm)''||''''||''18??''||''18''||''18''|| ||''H_laser_up (cm)''||''''||''14.6''||''12.8''||''14.8''||''''|| ||''H_laser_down (cm)''||''''||''''||''''||''''||''''|| ||''H_coast (cm)''||'''''''||''18??''||''18''||''18''|| ||''H_laser_up (cm)''||'''''''||''14.6''||''12.8''||''14.8''||'|| ||''H_laser_down (cm)''||'''''''||'||'''''''||'||'''''''|| ''Samuel starts the day by calibrating ultrasound probes. He also changed on probe (Acq 3. Third from the wall). Values are stored in the file : calib probe 02/03. He different positions are : 0/-0.5/-1/+1/+0.5/0'' ||''Position''||''M2''||''M1''||''M0''||''M-1''||''M-2''|| ||''H_coast (cm)''||''''||''''||''19''||''18''|| ||''H_laser_up (cm)''||''''||''''||''15''||''14.3''||''''|| ||''H_laser_down (cm)''||''''||''''||''''||''''||''''|| ||''H_coast (cm)''||'''''''||'||''19''||''18''|| ||''H_laser_up (cm)''||'''''''||'||''15''||''14.3''||'''''''|| ||''H_laser_down (cm)''||'''''''||'||'''''''||'||'''''''|| ''EXP33 =>14h55 10.5 D (FPS 3) Problems with the camera PCO at the beginning. We need to compare the date of the first record with the trigger to obtain its time of start (~15h08).'' ||''Position''||''M2''||''M1''||''M0''||''M-1''||''M-2''|| ||''H_coast (cm)''||''''||''19.5''||''19.7''||''18.2''|| ||''H_laser_up (cm)''||''''||''17''||''16''||''15.5''||''''|| ||''H_laser_down (cm)''||''''||''''||''''||''''||''''|| ||''H_coast (cm)''||'''''''||''19.5''||''19.7''||''18.2''|| ||''H_laser_up (cm)''||'''''''||''17''||''16''||''15.5''||'|| ||''H_laser_down (cm)''||'''''''||'||'''''''||'||'''''''|| ''Lumenera Settings'' ||''Position''||''M2''||''M1''||''M0''||''M-1''||''M-2''|| ||''H_coast (cm)''||''''||''20''||''20''||''19''|| ||''H_laser_up (cm)''||''''||''19''||''17''||''16''||''''|| ||''H_laser_down (cm)''||''''||''''||''''||''''||''''|| ||''H_coast (cm)''||'''''''||''20''||''20''||''19''|| ||''H_laser_up (cm)''||'''''''||''19''||''17''||''16''||'|| ||''H_laser_down (cm)''||'''''''||'||'''''''||'||'''''''|| ''Lumenera Settings'' ||''''||''mean_dt[[BR]](ms)''||''mean_dx[[BR]](pix/mm)''||''Area[[BR]](pix/pix)''||''exposure[[BR]](ms)''||''contrast''||''brightness''|| ||'''''''||''mean_dt[[BR]](ms)''||''mean_dx[[BR]](pix/mm)''||''Area[[BR]](pix/pix)''||''exposure[[BR]](ms)''||''contrast''||''brightness''|| ||''EXP39''||''310''||''1.2''||''4000x1800''||''200''||''1.7''||''37''||''1-8''|| ||''EXP40''||''310''||''1.2''||''4000x1800''||''200''||''1.7''||''37''||''17''|| ||''Position''||''M2''||''M1''||''M0''||''M-1''||''M-2''|| ||''H_coast (cm)''||''''||''6.6''||''6.9''||''7''|| ||''H_laser_up (cm)''||''''||''4.6''||''2.1''||''4''||''''|| ||''H_laser_down (cm)''||''''||''''||''''||''''||''''|| ||''H_coast (cm)''||'''''''||''6.6''||''6.9''||''7''|| ||''H_laser_up (cm)''||'''''''||''4.6''||''2.1''||''4''||'|| ||''H_laser_down (cm)''||'''''''||'||'''''''||'||'''''''|| ''Lumenera Settings'' ||''''||''mean_dt[[BR]](ms)''||''mean_dx[[BR]](pix/mm)''||''Area[[BR]](pix/pix)''||''exposure[[BR]](ms)''||''contrast''||''brightness''|| ||''EXP44''||''''||''1.2''||''4000x1800''||''150''||''2.8''||''46''||''6-10''|| ||''EXP45''||''''||''1.2''||''4000x1800''||''''||''''||''''||''17''|| ||''EXP46[[BR]]''||''''||''1.2''||''4000x1800''||''''||''''||''''||''14''|| ||''EXP47''||''''||''1.2''||''4000x1800''||''200''||''2.8''||''46''||''14''|| ||'''''''||''mean_dt[[BR]](ms)''||''mean_dx[[BR]](pix/mm)''||''Area[[BR]](pix/pix)''||''exposure[[BR]](ms)''||''contrast''||''brightness''|| ||''EXP44''||'''''''||''1.2''||''4000x1800''||''150''||''2.8''||''46''||''6-10''|| ||''EXP45''||'''''''||''1.2''||''4000x1800''||'||'''''''||'||''17''|| ||''EXP46[[BR]]''||'''''''||''1.2''||''4000x1800''||'||'''''''||'||''14''|| ||''EXP47''||'''''''||''1.2''||''4000x1800''||''200''||''2.8''||''46''||''14''|| ''EXP44 =>08h48 fin 9h28''